The X-ray machine is a device that produces X-rays. It is mainly composed of an X-ray tube, an X-ray machine power supply, and a control circuit. The X-ray tube assembly consists of a cathode filament (Cathod), an anode target (Anode), and a vacuum glass tube. composition:
1. Open the anode first, so that you can see that there is a wire passing through the part of the cable sleeve, first remove this wire.
2, and then open the cover of the cathode of the bulb.
3, then remove the 3 lines, big focus, small focus, common line
4, and then fix itX-ray tube assemblyOpen the 3 screws of the tube, and open the 3 screws fixed to the tube
5. Now the circular ring can be taken out and the X-ray tube can be removed
6. At this time, use one hand to reach in and hold the X-ray tube firmly, and the other hand can stabilize the tube. The inner hand should be forced to face the anode and turn counterclockwise, so that the X-ray tube can be taken off. .
7. When installing, be careful, follow this order of thinking, and also pay attention to the center point of the X-ray tube assembly to the center of the window.
8. Finally, after everything is fixed, you can add the burnt transformer oil to the tube (below 60 degrees Celsius)
- Schematic diagram of the basic structure of the applicable CT tube
- Development trends of the global CT tube production industry
- The next outlet in the medical industry: tube service for medical equipment
- X-ray tube production imaging principle
- CT Replacement Tube Independent Manufacturer
- Two major directions for the development of CT substitute tube technology and the market
- There are these methods for tube development to prolong the life of CT tube
- Warm Invitation|Konason & Shanghai Pharmaceuticals Sonnick will jointly launch tube products at the Shanghai Medical Expo
- Tube sales talk about CT maintenance common fault maintenance methods
- The anti-epidemic material CT tube in the unexpected reason
- The development history of CT tube